FPD-8010E provides a wide range of effective presentation and report functions utilizing FPD-8010E software and a scanner to digitize Prescale outputs to allow the user to produce multi-faceted measurement data such as pressure distribution and enlargement, cross-sectional distribution, and 3-D image display.
This serves to accelerate product development and improved yields based on detailed multi-faceted measurement data and analysis.
Converts Prescale density values into pressure values.
This manuscript hold-down scanner cover improves date-read precision.
The calibration sheet limits scanner-read errors to a fixed range.
Various data such as average pressure and maximum pressure are displayed.
The specified field is enlarged. （X 4, X 8, X 16）
Pressure distribution on a line passing through a specified point is shown on a line graph.
Pressure is displayed in 3-D format.
The upper and left segments of the total pressure are displayed on a bar graph.
Pressure on the circumference is displayed as a histogram.
Pressure data is exported to a text file.
Step-by-step pressure values are displayed in animated format.
FUJIFILM PRESSURE DISTRIBUTION MAPPING SYSTEM for PRESCALE
Prescale image input function
Single Read: 297mm X 210mm（11.7 in X 41.3 in）
0.125（200dpi）, 0.25（100dpi）, 0.5,1,2mm sq.
|Dedicated Cover Weight||
|Dedicated Cover Dimensions||
70（H） X 290（W） X 364（D）mm
Dedicated software, dedicated cover, calibration sheet, installation manual, software license
Epson V100, V200, V350, V300, V330, V33, V370 and V37
Windows XP, Vista and 7 (32/64bit)
Pentium® III 1GHz or Higher
512MB or more
XGA or better, 65,000 colors or more
A.Digital data saving is recommended because Prescale will loss its accuracy as certain elements pose adverse affects.
A.They will upgrade software elements.
A.No, the FPD-8010E is configured and guaranteed to operate with the recommended scanners only.
A.No, there is no need for regular maintenance. The FPD-8010E Series conduct calibration when you power up the system to check the characteristics change and deterioration of the scanner.
A.No. The FPD-8010E Series data can be utilized to prepare your own software run on other PCs. The information on structure of MPA file is disclosed for this purpose. Additionally, the components are required to calibrate and operate the system.
This document explains how to install the Update Module of FPD-8010E for Windows or Vista Operating Systems.
Double-click the following icon with the mouse to execute. (Go to 2)
The old file is deleted after decompressing the program.
Click the “Execute” button to update the program.
Click the “Yes” button. (Go to 7)
(Go to 8 if files are not found.)
Click the “OK” button to complete the update. The old file will be deleted. (Go to 10)
The following should be the main two reasons why the files are not found:
Click the “OK” button to select the destination where the program file downloaded.
Select the executive file and click the “Open” button.
Go to 6 after finishing successfully.
Go to 9 if the selected file is not FPD8010Win.exe.
The following dialog will show after setting the scanning condition if the update was not completed correctly or the SP curve file is broken.
Scanning is not possible and only the “Cancel” button is accepted.
Re-install FPD-8010E is necessary if re-update cannot solve the problem.