The content on this page is intended to healthcare professionals and equivalents.
High quality images for easier diagnosis
Tomosynthesis: making it possible to observe the internal structure of the breast
In breast tomosynthesis, the X-ray tube moves through an arc while acquiring a series of low-dose X-ray images.
The images taken from different angles are reconstructed into a range of Tomosynthesis slices where the structure of interest is always in focus.
The reconstructed tomographic images make it easier to identify lesions which might be difficult to visualize in routine mammography because of the presence of overlapping breast structures.
S-View(synthesized 2D image)function is available
Tomosynthesis by AMULET Innovality automatically produces not only tomograms obtained at 1 mm intervals but also a two- dimensional S-View image combining multiple slice images.
With the S-View image showing the overall view added to tomograms offering the views in detail, comprehensive image reading is possible.
ISR (Iterative Super Reconstruction)
The tomosynthesis iterative super-resolution reconstruction (ISR) method is applied to optimize image quality, achieving significant X-ray dosage reduction.
1. Reducing graininess of image in low-dose tomography
The image patterns are recognized to selectively suppress the patterns that do not exist in human body architectures as noise, to reduce distractive noises in the event of low-dose tomography.
2. Suppressing interference of human body architectures at different depths (as illustrated on the right)
In the process of reconstructing the 3D breast architecture from multiple 2D images, calcification, mass, spicula, mammary gland and other signals that emerge from different depths in the breast architecture are selected off to reproduce the breast architecture at the focus depth with greater fidelity.
3. Restoring the fine-structure
Our super-resolution technology is introduced to restore the fine-structure of calcification and other phenomena, the visibility of which is impaired by the movement of the X-ray tube, to facilitate interpretation of tomosynthesis images.
Offers significantly lower doses than the conventional method
With combination of 2D and Tomosynthesis Dose of 2 or less mGy is available*2
Static face guard for Tomosynthesis imaging (Face Guard T Comfort)
Fixing the face guard to the device instead of the tube part eliminates movement of the face guard during Tomosynthesis imaging. It can also be used as-is for normal mammography imaging.
Shortens the imaging cycle with a fast display and reconstruction
After a shot, the next shot in either 2D or 3D can be started with a cycle time of approx. 15 seconds.