The content on this page is intended to healthcare professionals and equivalents.
X-Ray Dose reduction needs to be a priority
Improved access to medical examinations results in X-ray dose increase.
Improved access to medical examinations results increased X-ray dose to the population.
As CT scanning becomes more wide-spread, the propotion of X-ray dose exposure resulting from CT scans increases.
Contrast Media Dose reduction needs to be a priority
Contrast-induced renal failure is a major concern.
Image diagnosis using iodinated contrast media provides a lot of useful information, but using contrast media in patients with impaired kidney function risks contrast induced nephropathy (CIN).Therefore, guidelines for risk reduction have been issued by the European Society of Urogenital Radiology, ESUR, and the American College of Radiology, ACR.
Challenge: Achieving high image quality at a low radiation dose
Measures are being taken worldwide to mitigate the exposed dose as the number of examinations increases. For example, the Diagnostic reference level (DRL), which is a measure of the radiation dose for radiological diagnosis, has been introduced in many countries, defining standards to promote scanning at the lowest possible dose consistent with maintaining high image quality.
Diagnostic reference level (DRL)
1. CTDIvol [mGy]
2. DLP [mGy・cm]
Challenge: Decreasing noise during low-tube-voltage scanning
To reduce the Contrast Media Nephropathy (CIN) risk, efforts have been made to reduce the amount of contrast media used during low tube voltage scanning.
However, low-tube-voltage scanning increases image noise and the choice of exposure factors become more crucial.
Basic properties of x-ray
1. SD variation of tube voltage
2. Adjustment of tube current
Our Iterative processing Intelli IPV delivers a Clear Image with Low Dose
Vision Model Technology
- The Physics Model reduces noise present in the FBP by adjusting the texture at a uniform rate, from high frequency to low frequency, and achieves a texture close to FBP.
- Statistical Model statistically reduces both the X-ray noise and electrical noise.
- The Object Model keeps the structure in consideration of the shape, size, contrast, position, etc. of the structure by modeling the change of morphological information.
Iterative processing Intelli IPV maintains normal image texture, even at a high dose reduction ratio, as well as reducing the noise, resulting in improved visibility. Vision Model Technology makes this possible. The technology controls image noise and quality by iterative arithmetic processing based on a statistical model, an object model and a physics model.
Conventional IR compared to FBP image
- The texture of the image changes
- Low contrast detectability is reduced
Main advantages of Intelli IPV
Intelli IPV achieves an image texture similar to FBP images by making the NPS (noise power spectrum) close to that of the FBP, as well as improving the low contrast detectability by up to 2 times.
Can image quality be adjusted according to the anatomical region and the pathology scanned?
Yes. The noise reduction effect has up to 9 selectable intensity levels (Standard: Lv.1〜4 and Strong: Lv.1〜5) for each anatomical region.
Is the reconstruction time fast enough to be used for routine examinations?
Yes, the reconstruction time is fast enough to be used for routine examinations.
What are the main advantages of Intelli IPV compared to FBP?
Achieves image noise reduction up to 90%, radiation dose reduction up to 83%, and improves low contrast detection capability up to 2 times.